The EnSite Precision™ Cardiac Mapping System allows for a high level of automation, flexibility and precision to help you diagnose a wide range of arrhythmias.1,2*
The EnSite Precision Cardiac Mapping System is designed to:
The EnSite Precision™ Cardiac Mapping System is a suggested diagnostic tool in patients for whom electrophysiology studies have been indicated. The EnSite Precision™ System interfaces to either the MediGuide™ Technology System or the EnSite Precision™ Module to combine and display magnetic processed patient positioning and navigation mapping information. When used with the EnSite™ Array™ Catheter, the EnSite Precision™ Cardiac Mapping System is intended to be used in the right atrium of patients with complex arrhythmias that may be difficult to identify using conventional mapping systems alone. OR, when used with an EnSite Precision™ Surface Electrode Kit, the EnSite Precision™ Cardiac Mapping System is intended to display the position of conventional electrophysiology (EP) catheters in the heart.
Journal articles, abstracts and other clinical evidence have analyzed the tracking and navigation accuracy, point collection, techniques and speed of the EnSite Precision Cardiac Mapping System. For a deep dive into the clinical data, visit the Journal Article information.
Performance bench testing was conducted to verify the tracking and navigation accuracy of the magnetic navigation and visualization technology in the EnSite Precision Cardiac Mapping System. This study tested the tracking accuracy and the navigation accuracy of the EnSite™ NavX™ Sensor Enabled™ technology in the EnSite Precision module with the FlexAbility™ ablation catheter, Sensor Enabled™ and the Advisor™ FL circular mapping catheter, Sensor Enabled™.
Table 1: Accuracy of EnSite Precision Cardiac Mapping System (v2.0.1) with EnSite NavX Navigation and Visualization Technology, Sensor Enabled Magnetic Field Scaling
|Sensor Enabled™ Catheters Tested||Tracking Accuracy Error (No. Samples)||Navigation Accuracy Error (No. Samples)|
|Advisor™ FL Circular Mapping|
Catheter, Sensor Enabled™
and Flexability Catheter,
(DWL Phantom Model)
Mean: 0.44 ± 0.45 mm
Mean: 0.18 ± 0.07mm
Max: 0.38 mm
Mean: 0.45 ± 0.49 mm
Mean: 0.34 ± 0.16 mm
Max: .092 mm
* The open-platform feature of the EnSite Precision™ cardiac mapping system allows for use of almost any catheter for mapping, thus offering superior flexibility as compared to the CARTO‡ system by Biosense Webster, which limits use to Biosense Webster catheters only.
** In accordance with catheter indication for use.
You are about to enter an Abbott country- or region-specific website.
Please be aware that the website you have requested is intended for the residents of a particular country or countries, as noted on that site. As a result, the site may contain information on pharmaceuticals, medical devices and other products or uses of those products that are not approved in other countries or regions
Do you wish to continue and enter this website?