To prep the device, verify Marker Lumen patency by flushing it with saline until saline exits the Marker Port.
View the animation library for System Overview, Device Preparation, and full deployment steps for Single Device Deployment and Multiple Device Deployment techniques.
Refer to Instructions for Use for additional information.
Shallow Deployment Angle
45º Deployment Angle
Before attempting to remove the device:
Difficult or unable to lower the Lever (close the Foot) and remove device after the Suture Deployment
Lack of hemostasis may be a result of incorrect order of knot advancement or tangled sutures.
You are about to enter an Abbott country- or region-specific website.
Please be aware that the website you have requested is intended for the residents of a particular country or countries, as noted on that site. As a result, the site may contain information on pharmaceuticals, medical devices and other products or uses of those products that are not approved in other countries or regions
Do you wish to continue and enter this website?