The Entrant™ ICD is MR Conditional when used with MR Conditional leads for full body scans using a 1.5T or 3T field strength MRI scanner.
For additional information about specific MR Conditional devices and leads, including scan parameters, warnings, precautions, adverse conditions to MRI scanning and potential adverse events, please refer to the Abbott MRI Ready Systems Manual.
|Lead Model||Magnet (Tesla)||RF Transmit Conditions||Scan Region|
|Durata™ Defibrillation Lead|
7120Q (lead lengths: 58, 65 cm)
7122Q (lead lengths: 58, 65 cm)
|1.5T / 3T||Normal Operating Mode||Full-body|
|Optisure™ Defibrillation Lead|
LDA220Q (lead lengths: 58, 65 cm)
LDA210Q (lead lengths: 58, 65 cm)
|1.5T / 3T|
|Tendril™ STS Pacing Lead|
2088TC (lead lengths: 46, 52 cm)
|1.5T / 3T|
|Tendril MRI™ Pacing Lead|
LPA1200M (lead lengths: 46, 52 cm)
|MRI Mode||DOO; VOO; AOO; Pacing Off|
|MRI Base Rate||30-100 bpm|
|MRI Paced AV Delay||25-120 ms|
|MRI Pulse Amplitude||5.0 or 7.5 V|
|MRI Pulse Width||1.0 ms|
|MRI Pulse Configuration||Bipolar|
|MRI Timeout||Off; 3; 6; 9; 12; 24 hours|
*DOO, AOO and paced AV delays are only applicable to the DR models.