The Gallant™ ICD is MR Conditional when used with MR Conditional leads for full body scans using a 1.5T or 3T field strength MRI scanner.
For additional information about specific MR Conditional devices and leads, including scan parameters, warnings, precautions, adverse conditions to MRI scanning and potential adverse events, please refer to the Abbott MRI Ready Systems Manual.
|Lead Model||Magnet (Tesla)||RF Transmit Conditions||Scan Region|
|Durata™ ;Defibrillation ;Lead|
7120Q ;(lead ;lengths: ;58, ;65 ;cm)
7122Q (lead lengths: 58, 65 cm)
|1.5T / 3T||Normal Operating Mode||Full-body|
|Optisure™ Defibrillation Lead|
LDA220Q ;(lead ;lengths: ;58, ;65 ;cm)
LDA210Q (lead lengths: 58, 65 cm)
|1.5T / 3T|
|Tendril™ ;STS ;Pacing ;Lead|
2088TC (lead lengths: 46, 52 cm)
|1.5T / 3T|
|Tendril MRI™ Pacing Lead|
LPA1200M ;(lead ;lengths: ;46, ;52 ;cm)
|MRI Mode||DOO; VOO; AOO; Pacing Off|
|MRI Base Rate||30-100 bpm|
|MRI Paced AV Delay||25-120 ms|
|MRI Pulse Amplitude||5.0 or 7.5 V|
|MRI Pulse Width||1.0 ms|
|MRI Pulse Configuration||Bipolar|
|MRI Timeout||Off; 3; 6; 9; 12; 24 hours|
*DOO, AOO and paced AV delays are only applicable to the DR models.
You are about to enter an Abbott country- or region-specific website.
Please be aware that the website you have requested is intended for the residents of a particular country or countries, as noted on that site. As a result, the site may contain information on pharmaceuticals, medical devices and other products or uses of those products that are not approved in other countries or regions
Do you wish to continue and enter this website?